**Research article**| 17 Nov 2022

# The measurement of mean wind, variances, and covariances from an instrumented mobile car in a rural environment

Stefan J. Miller and Mark Gordon

**Stefan J. Miller and Mark Gordon**Stefan J. Miller and Mark Gordon

- Department of Earth and Space Science and Engineering, York University, Toronto ON, M3J 1P3, Canada

- Department of Earth and Space Science and Engineering, York University, Toronto ON, M3J 1P3, Canada

**Correspondence**: Stefan J. Miller (sjmiller@yorku.ca)

**Correspondence**: Stefan J. Miller (sjmiller@yorku.ca)

Received: 18 May 2022 – Discussion started: 13 Jun 2022 – Revised: 27 Aug 2022 – Accepted: 21 Sep 2022 – Published: 17 Nov 2022

On 20 and 22 August 2019, a small tripod was outfitted with a sonic anemometer and placed in a highway shoulder to compare with measurements made on an instrumented car as it traveled past the tripod. The rural measurement site in this investigation was selected so that the instrumented car traveled past many upwind surface obstructions and experienced the occasional passing vehicle. To obtain an accurate mean wind speed and mean wind direction on a moving car, it is necessary to correct for flow distortion and remove the vehicle speed from the measured velocity component parallel to vehicle motion (for straight-line motion). In this study, the velocity variances and turbulent fluxes measured by the car are calculated using two approaches: (1) eddy covariance and (2) wavelet analysis. The results show that wavelet analysis can better resolve low frequency contributions, and this leads to a reduction in the horizontal velocity variances measured on the car, giving a better estimate for some measurement averages when compared to the tripod. A wavelet-based approach to remove the effects of sporadic passing traffic is developed and applied to a measurement period during which a heavy-duty truck passes in the opposite highway lane; removing the times with traffic in this measurement period gives a reduction of approximately 10 % in the turbulent kinetic energy. The vertical velocity variance and vertical turbulent heat flux measured on the car are biased low compared to the tripod. This low bias may be related to a mismatch in the flux footprint of the car versus the tripod or perhaps to rapid flow distortion at the measurement location on the car. When random measurement uncertainty is considered, the vertical momentum flux is found to be consistent with the tripod in the 95 % confidence interval and statistically different than 0 for most measurement periods.

- Article
(7804 KB) -
Supplement
(1426 KB) - BibTeX
- EndNote

Measurements of atmospheric means, variances, and covariances obtained from an instrumented mobile car can provide low-cost, in situ observations close to the ground and over a large measurement domain. Hereafter, “instrumented mobile car” refers to all potential on-road vehicles that could serve as a measurement platform, including cars, sport utility vehicles, pickup trucks, minivans, or larger mobile laboratories that use a heavy-duty truck. Previous investigations have largely used instrumented mobile cars for the measurement of near-surface atmospheric means, but minimal attention has been given to their use for the measurement of turbulence (i.e., variances and covariances). In the nocturnal boundary layer characterized by stable conditions and weak flow, turbulence near the surface mainly originates from poorly understood non-stationary mechanical shear and submesoscale motions (Mahrt et al., 2012; Van De Wiel et al., 2012), such as low-level jets, thermotopographic wind systems (i.e., katabatic flow), and breaking gravity waves (Salmond and McKendry, 2005). In the very-stable boundary layer, the generated turbulence is often intermittent and results in the vertical transport of scalars (i.e., heat, pollutants), but stationary towers may be too isolated and “site-specific” to adequately sample the temporally and spatially localized turbulence (Salmond and McKendry, 2005). The mobile car, however, can measure along a driven path, which may provide a more representative sample of turbulence near the surface compared to a stationary tower. In addition, the mobile car may also be used to obtain in situ wind and turbulence measurements near the surface within the urban boundary layer, measurements that may help validate high-resolution, street-level models. In the near-surface urban boundary layer, the strength of the wind and the intensity of turbulence are influenced by the composition of buildings and trees (Mochida et al., 2008; Gromke and Blocken, 2015; Hertwig et al., 2019; Krayenhoof et al., 2020) and can have a significant impact on pedestrian comfort (Hunt et al., 1976; Yu et al., 2020), and neighborhood-level pollutant dispersion (Aristodemou et al., 2018; Su et al., 2019). The mobile car involves fewer logistical limitations (i.e., permits, vandalism) and potentially affords a greater spatial coverage when compared to the installation of a stationary tower in a high-density urban area. Furthermore, as the resolution of numerical weather prediction models continues to improve, the measurement of localized variations in near-surface heat, momentum, and moisture fluxes may improve the prediction of convective storms (Markowski et al., 2019).

The instrumented mobile car has been used in various investigations to measure atmospheric means near the surface (Bogren and Gustavsson, 1991; Straka et al., 1996; Achberger and Bärring, 1999; Armi and Mayr, 2007; Mayr and Armi, 2008; Taylor et al., 2011; Smith et al., 2010; White, 2014; Curry et al., 2017; de Boer et al., 2021). Gordon et al. (2012) and Miller et al. (2019) used the instrumented car for the measurement of velocity variances on highways to quantify vehicle-induced turbulence. Despite the increasing number of investigations using instrumented mobile car systems for atmospheric measurements, there are limited studies that examine their performance and accuracy for the measurement of the mean flow, velocity variances, and covariances.

Achberger and Bärring (1999) investigated the accuracy of mean
temperature measurements made on a minibus in low-speed driving conditions
(8 to 11 m s^{−1}) by installing four thermocouples at various heights
(0.5, 1, 2, and 4 m). From their results, they developed a spectral
correction for the measured air temperature to remove the effects due to
thermal inertia of the thermocouples. More recently, Anderson et al. (2012)
evaluated the feasibility of using passenger vehicles (9 in total) to
collect mean air temperature and air pressure measurements on roads, with
the end goal of improving road weather forecasts to reduce weather-related
traffic fatalities. They found good agreement for mean air temperature
measurements made on passenger vehicles when compared to mean air
temperature measurements made by stationary weather stations; they also found poor
agreement for air pressure.

Belušic et al. (2014) is the first known study to evaluate a three-dimensional sonic anemometer (model CSAT3, sampling frequency of 20 Hz) affixed to a passenger vehicle for its accuracy at measuring atmospheric variances and covariances in addition to atmospheric means. In their setup, the sonic anemometer was supported by a sophisticated arm and lattice aluminum frame; the arm held the sonic above the vehicle's top at a height of 3 m from the ground, positioned slightly ahead of the vehicle's front end. Recently, Hanlon and Risk (2020) investigated how the placement of a sonic anemometer on the vehicle affects the accuracy of velocity measurements by applying computational fluid dynamics modeling in combination with mobile car measurements. The anemometers were placed vertically upward on top of the vehicle's roof.

If 1 min averages are assumed, then measurements (i.e., wind velocity, gas
concentration) obtained from an instrumented car traveling at near-highway
speeds (i.e., 15 to 25 m s^{−1}) are made over a significant spatial path
on the order of 10^{3} m, where surface variations (i.e., vegetation,
building structures, other traffic) can be significant. A single spatial
path measured by the vehicle may therefore feature flow conditions that are
not stationary and an upwind surface that is not homogenous. This calls into
question the applicability of the eddy covariance (EC) method, which
requires near-stationary conditions to reduce uncertainties in the
estimation of variances and covariances. During their investigation,
Belušic et al. (2014) made car measurements on a nearly flat, homogenous
portion of remote rural highway without traffic and without large upwind
obstacles, such as trees and houses. Therefore, their investigation
represented an “idealized” case. Even so, they found instances where the
car-measured horizontal velocity variances were significantly overestimated
compared to measurements made by a nearby stationary tower. They concluded
that non-stationarity of the flow was the likely cause leading to the
anomalously large car-measured horizontal velocity variances. Their results
demonstrate that non-stationarity of the flow cannot be ignored when
measuring on an instrumented mobile car. Recently, Schaller et al. (2017)
applied wavelet analysis as an alternative technique to estimate turbulent
methane fluxes measured by a fixed tower in non-stationary conditions. For
periods fulfilling the stationarity requirement, the wavelet flux was in
excellent agreement with eddy covariance flux, but for periods where the
stationarity requirement was violated, the wavelet flux was found to be more
reliable and provided a better estimate. Since their work, wavelet analysis
applied to analyze turbulent fluxes has become more common (von der Heyden
et al., 2018; Göckede et al., 2019; Conte et al., 2021).

The present work investigates an instrumented mobile car setup (shown in Fig. 1) by comparing car-based measurements with measurements made by a small roadside tripod. Our setup differs from Belušic et al. (2014) in two main ways, which are necessary to make the vehicle safe for on-road driving with other vehicles: (1) our sonic anemometer is held closer to the vehicle and situated over the vehicle's front end, and (2) the sonic anemometer is held closer to the ground at a height of 1.7 m, which is near the height of the vehicle's top. We selected this design to investigate whether the sonic anemometer can be held closer to the vehicle and still provide measurements that are representative of the mean flow and turbulence near the surface, allowing road-safe vehicle operation without compromising the measured data. While farmland is common in our measurement domain, the car also traveled past many large trees and houses and experienced the occasional passing vehicle traveling in the opposite direction. Therefore, we investigate if the mobile car measurements are still representative of the turbulence statistics near the surface in a less idealized case, where the upwind surface and terrain are not homogenous and where the measured flow is affected by many surface obstacles, including other traffic. Thus, this work aims to help design a low-cost experiment to measure and analyze on-road velocity variances and covariances using an instrumented car, in the presence of sporadic passing traffic and upwind surface inhomogeneities. This study investigates how these inhomogeneities affect the calculated statistics. Wavelet analysis is considered as an alternative technique to eddy covariance for the estimation of velocity variances and covariances measured on the car and is applied to quantify and remove the effects of sporadic passing traffic. The potential sources of measurement uncertainty on the car are quantified and discussed.

## 2.1 Instrumented car

A sport utility vehicle (SUV) was outfitted with instrumentation fastened to
the vehicle using a roof rack, as shown in Fig. 1. A 40 Hz,
three-dimensional sonic anemometer (Applied Technologies, Inc., model type
“A” or “Vx”) was installed on a support arm located at the front end of
the vehicle at a height of *z*_{m} = 1.7 m. Since the “A” type is rated
for higher flow velocities, once it became available for use it was
installed, and the “Vx” type was removed. This change was done to test
how the specific sonic anemometer model affects the measured velocities. The
“A”, “Vx”, and “V” type sonic anemometers (“V” is used on the roadside tripod) have an accuracy of ±0.1 m s^{−1} within a measurement range of ±60, ±20, and ±15 m s^{−1}, respectively. To limit the effect of
vibrations on the measurements made by the sonic anemometer, the horizontal
arm holding the anemometer was supported by two metal rods attached to the
vehicle's front end. The forward scene was recorded by a Thinkware F750
dashboard camera (30 frames per second), which encodes 1 Hz measurements of latitude,
longitude, and vehicle speed (*s*) as metadata in each MP4 file.

The coordinate system of the sonic anemometer on the car is defined
(assuming an observer is sitting inside of the vehicle facing toward the
front hood) so that measured velocity parallel to vehicle motion (*u*_{m})
is positive toward the car, the measured lateral velocity (*v*_{m}) is
positive toward the right, and the measured vertical velocity (*w*_{m}) is
positive upward. Subscript *m* denotes a raw measured value.

## 2.2 Roadside tripod

On 20 and 22 August 2019, a small tripod was assembled and placed at the
roadside (i.e., in the highway shoulder) to compare with measurements made
by the instrumented car as it traveled past the stationary tripod. The
tripod was equipped with a three-dimensional sonic anemometer (Applied
Technologies, Inc., model type “V”) that recorded at a frequency of either
10 Hz (20 August) or 20 Hz (22 August). Each day, the sonic was installed at a measurement height of *z*_{m} = 1.4 m. On 22 August, the tripod also had a
Thinkware X700 dashboard camera (30 frames per second) installed to record passing
traffic. To investigate the effect of tripod vibrations on the measurements,
we tied down the system with string on 22 August but left it free to
vibrate on 20 August.

## 2.3 Measurement site

The measurement site was agricultural fields located on either side of a
two-lane highway. The traffic on 22 August passing our measurement site was
more significant than on 20 August; the traffic composition on 22 August included
occasional large trucks, and we did not observe any large trucks passing our
measurement site on 20 August. Both days featured fair weather, with sky
conditions ranging from mainly sunny on 20 August to partly cloudy on 22 August.
The wind direction measured at nearby Egbert weather station (maintained by
Environment and Climate Change Canada, with measurements obtained at a height
of 10 m) ranged between 160 and 200^{∘} on 20 August and 310 and 340^{∘} on 22 August. The mean wind ranged between 4.2 and 5.6 m s^{−1} on 20 August and 3.8 and 5.0 m s^{−1} on 22 August. The Egbert weather station is located
about 16 km north of the measurement site.

The road is relatively flat near the tripod location, but in general, the
terrain is not flat and homogenous in this area. The study area (which spans
about 10 km) has several hills, with slopes up to 10^{∘}. The
elevation ranges between 200 and 300 m above mean sea level, and there are
areas with numerous trees and some structures located upwind of the highway.
The tripod was located at an elevation of 277 m on 20 August and at an
elevation of 222 m on 22 August (estimated from Google Earth). For reference, the Egbert weather station is at an elevation of 251 m.

In this work, a *measurement track* refers to the specific ground path driven by the vehicle, while a *measurement pass* refers to a specific set of measurements made on a particular
track. Each measurement pass can be further divided into “A” and “B”,
representing the specific direction driven by the vehicle on a particular
track. On each day, two different 1000 m tracks (Track #1 and Track #2) are chosen to compare with measurements made on the tripod. Track #1 is centered on the location of the tripod and consists of an equal
amount of highway on either side of the tripod (i.e., 500 m before the
tripod and 500 m after the tripod). Track #2, however, begins 120 m away
from the tripod and continues for 1000 m; thus, it does not include the
highway directly in front of the tripod. Track #1 and Track #2 (for
each day) are displayed in Fig. 2 as yellow and blue lines, respectively.
The location of the tripod in Fig. 2 is displayed as a marker with a star
enclosed. Track #1 and Track #2 are chosen to examine how the choice
of measurement track impacts the comparison of turbulence statistics between
the car and tripod. Track #1 and Track #2 overlap spatially for 380 m,
and so a portion of the data contained within both measurement tracks are
identical for each trip past the tripod. Table 1 gives the number of
measurement passes performed on each measurement track. The amount of
measurement passes that are excluded (from both Track #1 and Track #2)
due to traffic ahead of the instrumented car is also given. Two extra
measurement passes corresponding only to Track #2 were also analyzed on
22 August, where the car was parked at the tripod and then drove away (a
constant vehicle speed was achieved before 120 m). Since the car did not
travel down the entire length of measurement Track #1 prior to parking at the
roadside, there are no corresponding Track #1 for these two measurement
passes on Track #2.

## 2.4 Flow distortion and sensor corrections

Measurements made on an instrumented car may be significantly impacted by
flow distortion. Flow distortion originates from vehicle movement (speed
*s*) and from the ambient horizontal wind (*u*_{H}) that is present even when the vehicle is stationary; *u*_{H} may be at an angle to the vehicle,
potentially leading to flow distortion in both components of the measured
horizontal velocity (i.e., *u*_{m}, *v*_{m}). Further impacts on the
measurements can occur from sensor misalignment and sensor limitations that
occur while measuring in high flow velocities. Flow distortion at the
location of the sonic anemometer is investigated by analyzing measurement
passes that are separated into part A and B. A and B are each driven on the
same length of highway but in opposite directions (following Belušić et al.,
2014). Before investigating flow distortion, the sonic anemometer data are
filtered for spikes. Here, a spike is defined as an unrealistic sequence of 2
or less data points and is identified by applying a non-linear median
filter according to Starkenburg et al. (2016). For the measurements
considered in this paper, the effect of this spike removal on the calculated
statistics is minimal (i.e., in any measurement pass, there are 2 or less
flagged values). Measurements flagged as spikes are removed and replaced
with linearly interpolated values. If it is assumed that the mean ambient
vertical velocity $\stackrel{\mathrm{\u203e}}{w}$ ≈ 0 m s^{−1} and that the flow is
in steady state during A and B, with measurements made at a constant vehicle
speed *s*, then following Belušić et al. (2014) and Miller et al. (2019), we
can assume three relationships (here, an uppercase variable (*U*, *V*, *W*,
*S*) represents an averaged or binned value, while a lowercase variable
represents an individual measurement):

- i.
Without flow distortion, the average measured vertical velocity (

*W*) at any measured longitudinal velocity (*U*) is expected to be equal to 0 over a sufficiently long record. That is,*W*is not expected to have any dependence on*U*. However, in the presence of flow distortion on the mobile car,*W*becomes a function of*U*. - ii.
The average velocity recorded over both travel directions (

*U*_{AB}as a function of*S*) is expected to follow the relationship ${U}_{\mathrm{AB}}\left(S\right)=\mathrm{0.5}\left({U}_{\mathrm{A}}\left(S\right)+{U}_{\mathrm{B}}\left(S\right)\right)=S$, since any wind component parallel to the direction of vehicle motion is canceled out by traveling the same distance in both directions. - iii.
The lateral velocity

*V*measured over all of A and all of B is expected to follow the relationship ${V}_{\mathrm{AB}}=\mathrm{0.5}\left[{V}_{\mathrm{A}}+{V}_{\mathrm{B}}\right]=\mathrm{0}$, since the coordinate system rotates 180^{∘}when the vehicle changes direction.

Figure 3a shows *W* binned according to *U*, with binning completed using
a bin size of 1 m s^{−1}. The data shown in Fig. 3 includes all
back-and-forth passes completed on 20 and 22 August, and the binned data are
derived from individual measurements made by the 40 Hz sonic anemometer
(every 0.025 s). Each bin requires at least 80 independent samples (2 s of
data), otherwise it is rejected. Binning using individual measurements is
done instead of averaging over all of A and over all of B, since it is
difficult to maintain a constant vehicle speed during each part of the
measurement pass. However, most measurements of *U* fall into 2 to 4 speed
bins during a particular back-and-forth pass consisting of parts A and B.
The anemometer was not removed from the vehicle between 20 and 22 August;
therefore, the results should be consistent across both days. Figure 3a
demonstrates that flow distortion at the measurement location is significant
in this study, and *W* increases linearly with increasing *U* (coefficient
of determination, *R*^{2} = 0.99). The measured velocity field is
corrected by applying a coordinate rotation to give a 0 mean vertical
velocity (assuming there is no flow distortion effect in *v*_{m}), as

Here, *θ* is set to the median of *θ*_{b}, where ${\mathit{\theta}}_{b}=\mathrm{atan}\left({W}_{b}/{U}_{b}\right)$ and subscript *b* represents individual
binned values of 1 m s^{−1} size (i.e., from Fig. 3a). *θ*_{b}
does not show any dependence on *U* for the vehicle speeds investigated in
this study (i.e., for *S* > 15 m s^{−1}; see Fig. S1 in the
Supplement). For the data shown in Fig. 3a, *θ* = 7.54^{∘} (interquartile range of 0.32^{∘}).

Figure 3b shows *U*_{1} binned according to *S*. In Fig. 3b, at *S* > 17 m s^{−1}, the results suggest that *U*_{1} is
overestimated. The same analysis performed on 30 August did not show this
overestimation in *U*_{1} for higher *S* (Fig. 4b); however, the setup on 30 August used a sonic anemometer that is rated for higher flow velocities up to 60 m s^{−1} (Applied Technologies, model “A”). This suggests that the overestimation in *U*_{1} on 20 and 22 August for *S* > 17 m s^{−1} is likely an instrument-related limitation rather than a direct
effect of flow distortion. Taking the difference between the least-square
fit and the expected relationship (i.e., *U*_{1}=*S*), the overestimation in
*u*_{1} (i.e., after applying Eq. 1) is

The overestimation, *u*_{excess}(*s*) is then removed from *u*_{1} to give *u*_{c}, as

No corrections are applied to *v*_{m}, since there is no clear relationship
with any measured variable (i.e., *U*, *S*; see Fig. S2). The corrections
outlined in Eqs. (1) through (3) are applied to all vehicle measurements
from 20 and 22 August, for which *s* > 0 m s^{−1}. After
correction for flow distortion, the 1 Hz vehicle speed is linearly
interpolated to 40 Hz and then removed from *u*_{c} to give the
meteorological wind speed component parallel to the direction of motion, as
(Belušić et al., 2014)

## 2.5 Wavelet analysis and the quantification of sporadic passing traffic

The continuous wavelet transform of a discrete time series *x* containing
*N* data points, measured at a time step Δ*t*, is calculated as
(Torrence and Compo, 1998a)

The wavelet coefficients are calculated as the convolution of *x*, with a
dilated (*a*) and translated (*n*) wavelet function *ψ*_{0}, where *a*
is referred to as the wavelet scale and *n* is a localized time (position)
index. If *ψ* is complex, then the complex conjugate (∗) is used
to calculate ${G}_{n}^{x}\left(a\right)$. Following Torrence and Compo (1998a), the analyzing wavelet is normalized to have unit energy, so that

where *ψ*, in this work, is the complex Morlet wavelet,

The Morlet wavelet is chosen, since it has been shown to be well suited for
the analysis of atmospheric turbulence (Strunin and Hiyama, 2004; Salmond,
2005; Schaller et al., 2017). The total energy (or wavelet variance) of the
entire time series is preserved in the wavelet transform and can be
recovered by summing the scale-averaged wavelet power over all scales (*j*)
and times (*n*):

where Δ*j* = 0.25 determines the spacing between discrete
scales *a*_{j}=*a*_{0}2^{jΔj} (*a*_{0}=2Δ*t*) and *C*_{δ}=0.776 is a wavelet-specific reconstruction factor for
the Morlet wavelet. The Morlet wavelet scale can be converted to an
equivalent Fourier scale (i.e., period), as ${\stackrel{\mathrm{\u0303}}{a}}_{j}=\mathrm{1.03}{a}_{j}$. Like
the wavelet variance, given time series *x*_{n} and *y*_{n}, the wavelet
covariance (or turbulent flux) can be calculated as

where the real part (ℜ) of the wavelet cross-spectrum defines the wavelet
co-spectrum, and the imaginary part gives the wavelet quadrature spectrum
(Strunin and Hiyama, 2004; Paterna et al., 2016). For a 1000 m track
consisting of ${N}_{T}={T}_{m}/\mathrm{\Delta}t$ measurements (*T*_{m}∈*Z* is the integer second
length of the track), the wavelet variance, including timescales
up to index *a*^{∗}, can be calculated as

In Eq. (10), index value *a*^{∗} represents the maximum (Fourier
equivalent) wavelet scale and controls the timescales that are included in
the wavelet variance, which, in this work, is set to match *T*_{m} as closely
as possible. ${G}_{n}^{x}\left({a}_{j}\right)$ is calculated from a measured
time series with a temporal length of 11*T*_{m}, where the data
corresponding to the measurement pass (over which ${\mathit{\sigma}}_{{x}_{\mathrm{1}\phantom{\rule{0.125em}{0ex}}\mathrm{km}}}^{\mathrm{2}}$ is calculated) are located at the center of this
period (i.e., from $\mathrm{5}\le {T}_{m}<\mathrm{6}$). This approach is applied to ensure that
the wavelet transform coefficients used to calculate the wavelet variances
are not impacted by edge effects for scales up to *a*^{∗} (i.e., they do
not lie outside of the cone of influence) while still retaining good
computational efficiency (Torrence and Compo, 1998a; Schaller et al., 2017). Torrence and Compo (1998a) recommend zero padding a finite series of length
*T*_{m} to reduce edge effects, but in this study, there is no need to pad
the time series before or after the measurement pass with zeros, since the
instrumented car continued driving down the same road after measuring on
Track #1 and Track #2, providing continuous measured data before and
after each measurement pass. These continuous data limit edge effects in the
wavelet variances and covariances calculated over $\mathrm{5}\le {T}_{m}<\mathrm{6}$ for
${a}^{\ast}\le {T}_{m}$, providing a more reliable estimate for each
measurement pass. Hence, the additional data before and after each
measurement pass (equivalent to a spatial distance of about 10 km) come
from continuous driving in the vicinity of the tripod at a relatively
constant speed and, in most cases, on the same road. The instrumented car did
not come to rest, except briefly at a stop sign or to reverse direction.
There are two exceptions for measurement passes on Track #2, where the
car initially started from rest and reached a constant speed before
traveling 120 m from the tripod. Based on the cone of influence definition
by Torrence and Compo (1998a), wavelet coefficients for each measurement pass
are primarily influenced by data between $\mathrm{3.63}\le {T}_{m}<\mathrm{7.37}$ for ${a}^{\ast}\le {T}_{m}$. Therefore, the data between $\mathrm{0}\le {T}_{m}<\mathrm{3.63}$ and $\mathrm{7.37}\le {T}_{m}<\mathrm{11}$ have little impact on the calculated wavelet variance or
covariance and thus are not necessary to give a reliable estimate for the
measurement pass.

${\mathit{\sigma}}_{{x}_{\mathrm{1}\phantom{\rule{0.125em}{0ex}}\mathrm{km}}}^{\mathrm{2}}$ can be decomposed to give the wavelet
variance for each second of the track (likewise with scales up to index
*a*^{∗}), as

where *i*=0, 1, …, *T*_{m}−1, $N=\mathrm{1}/\mathrm{\Delta}t$ and

The wavelet variance calculated for each second allows the effects of sporadic passing traffic to be removed by excluding times when traffic is likely affecting the measurements made on the car (as determined by manual inspection of the video recordings), calculated as

where

and ${T}_{f}=\sum _{i=\mathrm{0}}^{{T}_{m}-\mathrm{1}}{\mathit{\delta}}_{i}$.

Using the real part of the wavelet coefficients, the original time series
*x* can be reconstructed at each *n*. By limiting the scales (for example,
selecting scales $j={J}_{min}$ to $j={J}_{max}$), a
wavelet-filtered time series can be constructed at each *n*, as

where $\mathit{\psi}\left(\mathrm{0}\right)={\mathit{\pi}}^{-\mathrm{0.25}}$ for the Morlet wavelet. Calculation of the wavelet transform is computationally intensive when Eq. (5) is used. By applying the convolution theorem, the wavelet transform can be completed much faster in Fourier space, and this approach is used here; the software developed to perform the continuous wavelet transform has been converted to IGOR Pro from Matlab code, available online by Torrence and Compo (1998).

## 2.6 Coordinate rotation

To compare the measurements made on the tripod to those made on the car, the
coordinate systems must be consistent. The initial step is to rotate the
individual measurements made on the vehicle into a meteorological coordinate
system (i.e., *u*_{met} positive toward the east and *v*_{met} positive
toward the north) using the vehicle's heading. This rotation is necessary,
since the vehicle's heading may change along the measurement path, leading
to a varying sonic anemometer coordinate system along a driven path. For
driven paths with large curvature, not performing the transformation to
meteorological coordinates gives incorrect mean values (and variances) that
are used to determine the rotation angles needed for transformation into a
streamwise coordinate system. For the highways investigated in this study,
the vehicle heading remains rather consistent over their length; hence, our
analysis only applies to straight vehicle motion, and we do not determine
uncertainties due to measurements through road curvature.

After rotation into meteorological coordinates, each track (on the car and tripod) is then rotated into a mean streamwise coordinate system following Wilczak et al. (2001), where $\stackrel{\mathrm{\u203e}}{u}$ is the mean wind and $\stackrel{\mathrm{\u203e}}{v}=\stackrel{\mathrm{\u203e}}{w}=\mathrm{0}$. The wavelet variances and covariances are likewise rotated into mean streamwise coordinates (unless otherwise indicated) using the same rotation angles applied to rotate the eddy covariance results.

## 2.7 Sampling errors

### 2.7.1 Random measurement uncertainty

For the calculation of turbulence statistics, the use of a finite record length gives rise to a random measurement uncertainty, since the record will not contain enough independent samples to accurately represent the ensemble mean (Lenschow et al., 1994). Further random measurement uncertainty can be introduced by non-stationarity in the record and white noise in the measured signal (Rannik et al., 2016). In this work, the magnitude of the random measurement uncertainty is estimated using two methodologies. All uncertainty estimations are after correction for flow distortion and rotation into a streamwise coordinate system. The first method, developed by Mann and Lenschow (1994), can be defined as

with the integral timescale (*I*_{ws}) calculated as

*I*_{wq} is estimated by numerically integrating the autocorrelation function
to the first zero crossing. In Eq. (15), ${z}^{\ast}\cong \mathrm{0}$ near the surface,
${r}_{wq}=\frac{\stackrel{\mathrm{\u203e}}{{w}^{\prime}{q}^{\prime}}}{{\mathit{\sigma}}_{w}{\mathit{\sigma}}_{q}}$ is the correlation
coefficient between *w* and *q*, and *T*_{m} is the averaging period (in
seconds) over which the covariance is calculated. For neutral stability,
*I*_{wq} can be approximated as $z/s$ (Finkelstein and Sims, 2001). For a
vehicle with a measurement height of *z*_{m} = 1.7 m, a mean wind speed of $\stackrel{\mathrm{\u203e}}{u}\approx $ 2.5 m s^{−1}, and a constant vehicle speed of *s* = 25 m s^{−1}, the result is *I*_{wq} ≈ 0.07 s. For the stationary tripod (*s* = 0) at a slightly lower height of *z*_{m} = 1.4 m, *I*_{wq} = 0.56 s for the
same wind speed. For a covariance of scalar *q* with the vertical velocity
*w*, the instantaneous flux is calculated as ${\mathit{\phi}}^{\prime}={w}^{\prime}{q}^{\prime}=\left(w-\stackrel{\mathrm{\u203e}}{w}\right)\left(q-\stackrel{\mathrm{\u203e}}{q}\right)$, and *φ*^{′}is used to
estimate the autocorrelation function needed for calculation of the integral
timescale (*I*_{wq}) from Eq. (16) (Rannik et al., 2016). The instantaneous
flux is introduced, since the cross-correlation is an asymmetric function,
making it unsuitable for estimation of the *I*_{wq}.

The second methodology outlined in Finkelstein and Sims (2001) gives an
estimation of the variance of a covariance (*δ*_{FS}):

where *m* is the number of samples required to ensure the integral time
scale (ITS) is sufficiently captured. ${\widehat{\mathit{\gamma}}}_{w,w}\left(p\right)$ and
${\widehat{\mathit{\gamma}}}_{w,q}\left(p\right)$ are the unbiased autocovariance and
cross-covariance, respectively, expressed as

and

The value of *m* is determined by calculating *δ*_{FS} as a function of *m* and choosing the value at which *δ*_{FS} reaches a constant or
asymptotic value as *m* is further increased. For the roadside tripod, a
value of *m* = 300 s is determined, while for the vehicle measurements, *m* = 30 s (see Figs. S4 and S5)

For wavelet analysis, Eq. (14) is applied to generate a wavelet
reconstructed time series (*q*_{f} and *w*_{f}) for scales up to *a*^{∗}. Thus, the reconstructed time series will exclude low frequency
contributions attributed to wavelengths *λ* > 1000 m. The
reconstructed time series are then rotated into mean streamwise coordinates
and subsequently used in Eqs. (18) and (19) to estimate *δ*_{FS}
for the wavelet covariance (and likewise for wavelet variances).

### 2.7.2 Random measurement uncertainty due to instrument noise only

The sonic anemometer's signal may be impacted by white noise, a form of
random measurement uncertainty. Lenschow et al. (2000) consider a stationary
time series with its mean removed (i.e., *w*^{′}(*t*)) that is impacted by
(uncorrelated) white noise, *ϵ*(*t*), where the autocovariance
function is

Since *w*(*t*) and *ϵ*(*t*) are uncorrelated, *ϵ*(*t*) is present
only at 0 lag, and so $\stackrel{\mathrm{\u203e}}{w\mathit{\u03f5}}=\mathrm{0}$. Equation (20) then reduces to ${\mathit{\gamma}}_{w,w}\left(\mathit{\tau}\right)=\stackrel{\mathrm{\u203e}}{{w}^{\prime}{{w}^{\prime}}_{t+\mathit{\tau}}}$ with ${\mathit{\gamma}}_{w,w}\left(\mathrm{0}\right)=\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}+\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}$. Based on the inertial subrange theory by Kolmogorov, the autocovariance function is
expected to follow (Lenschow et al., 2000; Wulfmeyer et al., 2010; Bonin et
al., 2016):

where constant *C* is associated with turbulent eddy dissipation. To
estimate $\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}$, Eq. (21) is typically fit to the first 5
lags of the autocovariance function, corresponding to time lags of 0.1 to
0.5 s for a 10 Hz signal of a sonic anemometer (Rannik et al., 2016). For
Doppler lidar measurements of the vertical velocity in convective
conditions, Bonin et al. (2016) fit Eq. (21) to the autocovariance function
for time lags up to half the integral timescale (i.e., *τ*=0.5*I*_{ww}). The fit is then extrapolated back to 0 lag to give ${\mathit{\gamma}}_{w,w}(\to \mathrm{0})$, and the variance attributed to white noise in the measured
signal is then estimated as (Lenschow et al., 2000; Mauder et al., 2013)

Some authors report a poor fit to Eq. (21) and instead apply a linear fit
extrapolation back to 0 lag to determine Eq. (22) (Lenschow et al., 2000;
Mauder et al., 2013; Langford et al., 2015). For measurements obtained on
the tripod and instrumented car, a linear fit extrapolation in addition to
Eq. (21) are used to estimate $\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}$. For tripod
measurements, time lags up to 0.5 s are used to determine the fit, but for
the car traveling at vehicle speeds near 20 m s^{−1}, only the first 3
points (up to 0.075 s) of the autocovariance function are used. Equation (21) may lead to an extrapolated value at 0 lag larger than *γ*_{w,w}(0), which gives “negative” and thus undefined $\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}$. Bonin et
al. (2016) noted a similar finding in their investigation when fitting the
autocovariance function to Eq. (21) for the vertical velocity measured from
Doppler lidar. They hypothesize that the undefined $\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}$
occurs when the genuine white noise in the signal is minimal and the
smallest scales of turbulence remain unresolved. Therefore, when
$\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}$ is negative and undefined, we assume that the true
white noise is minimal and that $\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}$ ≈ 0. Thus, for the analysis herein $\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}=max\left(\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}},\mathrm{0}\right)$.

## 2.8 Comparison of mobile car measurements to tripod measurements

In this work, we follow the approach of Belušić et al. (2014) and select a
fixed ground path to investigate means, variances, and covariances on the
car. Two different fixed 1000 m ground paths (*L*) are considered, referred
to as Track #1 and Track #2, and these tracks are compared to
measurements made by the tripod (see Sect. 2.3).

The averaging period (*T*_{m}) on the car is set to the temporal length of
the 1000 m track for atmospheric means. For car-measured atmospheric
variances and covariances, *T*_{m} is calculated from Taylor's hypothesis (as
${T}_{m}=L/\stackrel{\mathrm{\u203e}}{u}$), with an *L* = 1000 m track length.
On the instrumented car, we have $\stackrel{\mathrm{\u203e}}{u}\cong s$, where *s* is the
near-constant vehicle speed over the 1000 m track, and therefore *T*_{m} is
equivalent to the time it takes for the car to travel 1000 m (for both
eddy covariance and wavelet analysis). For the car, any measurement pass
that follows closely behind a vehicle is excluded from the results. To
quantify a wavelet variance or covariance on the car, the maximum wavelet
timescale (*a*^{∗}) must be chosen. In this study, *a*^{∗} is set to
match *T*_{m} as closely as possible (i.e., the temporal length of the 1000 m track). This approach is used so that the wavelet variance (or covariance)
is directly comparable to eddy covariance, since both methodologies will
include the same timescales (*a*^{∗} controls the maximum timescale
included in the wavelet variance or covariance).

*T*_{m} on the tripod is set to 5 min for atmospheric means, but for
atmospheric variances and covariances, *T*_{m} varies depending on the mean
5 min wind speed measured by the tripod ($\stackrel{\mathrm{\u203e}}{u}$) according to Taylor's
frozen hypothesis, where *L* = 1000 m. For the two measurement days
investigated here, *T*_{m} on the tripod ranges between 5 and 8 min. For
consistency, the averaging period used for calculation of the tripod means,
variances, and covariances is centered on the time that the instrumented car
passes the tripod (for both Track #1 and Track #2). The choice of *L*
on the tripod is not trivial, since *L* should be determined by taking into
consideration the vehicle speed in addition to the mean ambient flow. Since
the mean ambient flow in this study was relatively weak
(∼ 2.5 m s^{−1}) and typically at an angle to the vehicle, we have
$\stackrel{\mathrm{\u203e}}{u}\cong s$ on the car, but in strong ambient flow $\stackrel{\mathrm{\u203e}}{u}\ne s$; Taylor's hypothesis would suggest a different *L* on the tripod to compare with the 1000 m track driven by the car. For example, if $\stackrel{\mathrm{\u203e}}{u}$ = 30 m s^{−1} on the car with *s* = 22 m s^{−1}, then 1000 m traveled by
the car would correspond to a distance of *L*_{s} = 1364 m traveled by an air parcel, and this distance should be used to determine *T*_{m} on the
tripod – that is, ${T}_{m}=\frac{{L}_{s}}{\stackrel{\mathrm{\u203e}}{u}}>\frac{\mathrm{1000}}{\stackrel{\mathrm{\u203e}}{u}}$. The averaging periods adopted in this study for each methodology (wavelet
analysis or eddy covariance) and measurement system (car or tripod) are
summarized in Table 2.

## 3.1 Mean wind speed and mean wind direction

Figure 5 shows a scatter plot of (a) the 5 min mean wind direction on the tripod compared to the mean wind direction measured on the mobile car, and (b) the 5 min mean wind speed measured on the tripod compared to the mean wind speed measured on the mobile car. The mean wind speed shown is after rotation into streamwise coordinates. The gray lines in Fig. 5 denote a specific percentage of the tripod measured value (i.e., 100 % gives a one-to-one relationship), and this convention is used in the figures that follow. The mean bias error, $\mathrm{MBE}=(\mathrm{1}/N)\sum _{i=\mathrm{1}}^{N}({M}_{\mathrm{c}}-{M}_{\mathrm{t}})$, and the root mean squared error, $\mathrm{RMSE}=(\mathrm{1}/N)\sum _{i=\mathrm{1}}^{N}\sqrt{{\left({M}_{\mathrm{c}}-{M}_{\mathrm{t}}\right)}^{\mathrm{2}}}$, are given in Table 3. Here, the subscripts c and t refer to the car and the tripod. The tripod is therefore used as a “ground truth” for the car measurements.

The mean wind speed shown in Fig. 5b shows relatively good agreement between the car and tripod, with no significant bias (${\mathrm{MBE}}_{\mathrm{car}}/{\stackrel{\mathrm{\u203e}}{u}}_{\mathrm{tripod}}$ = 2 % and
${\mathrm{RMSE}}_{\mathrm{car}}/{\stackrel{\mathrm{\u203e}}{u}}_{\mathrm{tripod}}$ = 22 %). When the
analysis is separated by tracks, the agreement is best for Track #1;
RMSE = 0.43 and 0.71 m s^{−1} for Track #1 and
Track #2, respectively (see Tables S1 and S2). If $\stackrel{\mathrm{\u203e}}{u}$ measured
on the tripod is used as a normalizing factor, the normalized root mean
squared error of $\stackrel{\mathrm{\u203e}}{u}$ (NRMSE) is 18 % and 30 % for Track #1 and Track #2, respectively. The mean wind direction on the car agrees well with the tripod on both Track #1 and Track #2, as shown in Fig. 5a, where most points fall within 20^{∘} of the
one-to-one line.

To investigate how the car performs for shorter averaging periods,
non-overlapping intervals of 10 s duration are examined on 20 and 22 August. There are 263 and 250 such intervals on 20 and 22 August, respectively, and these
represent times that the vehicle is driving in the vicinity of the tripod
(i.e., within about 10 km) and not necessarily on a 1000 m track. The
results are shown in Table 4, which displays the average meteorological wind
components (*u*_{met} and *v*_{met}), the mean wind direction, and the mean
wind speed (after rotation into streamwise coordinates). Statistics are also
shown in Table 4, including the median, maximum, and minimum values in each
set and the interquartile range (IQR). The standard deviation of the wind
direction is calculated using the Yamartino algorithm (Turner, 1986). The
results show that the wind direction is rather consistent on both days for a
shorter averaging period of 10 s, where the wind direction standard
deviation is 38^{∘} on 20 August and 31^{∘} on 22 August. While the
average of all 10 s mean wind speeds on 20 and 22 August is consistent with the measurement passes shown in Fig. 5b, there can be significant variation in each individual interval, as demonstrated by the large IQR and
maximum and minimum values (IQR = 1.30 and 1.86 m s^{−1} on 20
and 22 August respectively). This demonstrates that using short averaging
periods on the mobile car allows the measurement of localized flow variations,
where the magnitude of the flow may vary significantly but the direction
remains relatively constant in comparison.

## 3.2 Velocity variances and covariances

Figure 6 shows the velocity variances measured on the instrumented car
compared to the velocity variances measured on the tripod. The velocity
variances measured on the car are calculated using the typical statistical
approach, denoted as EC (i.e., for time series *x* with *N* points, ${\mathit{\sigma}}_{x}^{\mathrm{2}}=\left(\mathrm{1}/N\right){\sum}_{i=\mathrm{1}}^{N}{\left({x}_{i}-\stackrel{\mathrm{\u203e}}{x}\right)}^{\mathrm{2}}$) or wavelet analysis (i.e., Eq. 10). Only statistical
velocity variances measured by the tripod (and covariances calculated using
eddy covariance) are presented herein. For measurements made on the tripod,
the effect of applying wavelet analysis to calculate variances and
covariances is minimal compared to the instrumented car (see Fig. S3).
Furthermore, for some measurement passes, the Morlet wavelet applied to the
tripod suffers from edge effects that cannot be avoided, since the tripod
recordings were abruptly ended at the end of each measurement day. For
wavelet analysis, the maximum wavelet scale (index *a*^{∗}) is chosen to
correspond as closely as possible to the temporal length of the measurement
track to ensure that both calculation methods retain the same spatial scales
and are therefore comparable (see Sect. 2.8). For the car measurement tracks
investigated here, the temporal length ranges between 40 and 60 s, and all
measurement tracks have a maximum spatial scale of approximately 1000 m.

Applying wavelet analysis to estimate the horizontal velocity variances
leads to a significant reduction in the magnitude compared to EC for some
passes, specifically for those passes reporting the largest horizontal
velocity variances, as shown in Fig. 6a and b. This reduction results in
an improved agreement between the two measurement systems; for
$\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$, wavelet analysis gives RMSE_{W} = 0.75 m^{2} s^{−2} compared to RMSE_{EC} = 1.44 m^{2} s^{−2} for EC. However, retaining larger scales in the wavelet
variance calculation (i.e., corresponding to spatial scales exceeding 1000 m) gives horizontal velocity variances that are larger and more consistent
with EC. This suggests that, compared to EC, wavelet analysis can better resolve low
frequency variations occurring at spatial scales near and exceeding 1000 m. Low frequency contributions on the car may arise from
variation in the flow that results only from a changing upwind environment; therefore, this effect would not be captured by a stationary monitoring
station. As discussed in Sect. 2.5, wavelet analysis is applied to a time
series with a temporal length 11 times longer than the time series used to
calculate the EC variances, giving wavelet analysis superior low frequency
resolution compared to eddy covariance.

Despite the improved agreement when wavelet analysis is applied to estimate the horizontal velocity variances, there are still instances where $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{{v}^{\prime}}^{\mathrm{2}}}$ measured by the mobile car are larger than what is measured by the roadside tripod. Given the public highway where the study was conducted, some measurement passes inevitably have sporadic traffic that was traveling in the opposite direction as the mobile car (as determined by visual inspection of the video). The passing traffic can significantly impact the velocity variances measured on the car due to vehicle-induced turbulence, especially in the case of passing heavy-duty trucks (Gordon et al., 2012; Miller et al., 2019). For the measurement passes shown in Fig. 6, there are two instances where a heavy-duty truck traveled in the lane opposite to the instrumented car as well as a few occasions where passenger vehicles (i.e., SUV, cars) traveled past the car.

Figure 7 displays the 1 s wavelet variance calculated using Eq. (11) for
three different measurement passes from Track #2 (on 22 August); Fig. 7a
had two simultaneously passing sport utility vehicles (SUV), and Fig. 7c had a
passing heavy-duty truck followed in quick succession by an SUV. Wavelet
analysis is performed on the measured velocities in a meteorological
coordinate system (i.e., *u*_{met}, *v*_{met}), with *a*^{∗} extending
up the temporal length of the measurement pass (i.e., the same *a*^{∗}
used for the wavelet variances presented in Fig. 6). Each measurement pass
shown in Fig. 7 was performed in the same direction and in the highway lane
closest to the tripod (i.e., on the downwind side of the highway). Traffic
is denoted by a circled area in the respective figure panel. With these
instances of traffic included, the velocity variances are 1.68, 1.38, and 0.21 m^{2} s^{−2} for
$\stackrel{\mathrm{\u203e}}{{{u}_{\mathrm{met}}^{\prime}}^{\mathrm{2}}}$, $\stackrel{\mathrm{\u203e}}{{{v}_{\mathrm{met}}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{{w}_{\mathrm{c}}^{\prime}}^{\mathrm{2}}}$,
respectively. Removing the 1 s wavelet variances corresponding temporally
with these passing vehicles (9 s in total) gives a
$\stackrel{\mathrm{\u203e}}{{{u}_{\mathrm{met}}^{\prime}}^{\mathrm{2}}}$, $\stackrel{\mathrm{\u203e}}{{{v}_{\mathrm{met}}^{\prime}}^{\mathrm{2}}}$, and $\stackrel{\mathrm{\u203e}}{{{w}_{\mathrm{c}}^{\prime}}^{\mathrm{2}}}$ of 1.47, 1.29, and 0.17 m^{2} s^{−2},
respectively, representing about a 10 % reduction in the turbulent kinetic
energy during this measurement pass. This demonstrates that even limited
traffic traveling in the highway lane adjacent to the car (and in the
opposite direction) can substantially increase the magnitude of the velocity
variances measured by the car on a 1000 m track, especially heavy-duty
trucks. In Fig. 7a, two SUVs passed by the mobile car in quick succession,
but the passage of these vehicles is not discernable as a localized increase
of the 1 s wavelet variances. This suggests that the vehicle wakes did not
advect past the instrumented car during this measurement pass, and thus no
removal is warranted.

For the measurement pass shown in Fig. 7b, there is a noticeable increase in the 1 s horizontal velocity variances about 450 m into the measurement track. A similar trend is also seen in Fig. 7c. Before 450 m, there are many large trees and houses upwind of the highway, but after 450 m, the upwind environment becomes open farmland (i.e., limited obstructions to the mean flow). The presence of many trees and houses in close proximity acts as a windbreak, forcing the flow to accelerate and rise over the surface obstructions. The flow is reduced downwind of the surface obstruction (Taylor and Salmon, 1993; Mochida et al., 2008), and close to the surface just after the obstruction (i.e., the near wake) is the “quiet zone”, where the horizontal velocity variances are reduced in comparison with the undisturbed upwind flow (Lee and Lee, 2012; Lyu et al., 2020). Therefore, the reduced horizontal velocity variances for the first few hundred meters of the track may be related to the quiet zone generated by the many trees and houses upwind of the road. After about 450 m, the upwind environment becomes relatively open, and the flow measured on the car increases, with this increase continuing over the remainder of the track. The changing wind speed along the track introduces a trend in the horizontal velocity record measured on the car.

On the instrumented car, $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ is biased low by 30 % to 50 % (${\mathrm{MBE}}_{\mathrm{EC}}=-\mathrm{0.11}$ m^{2} s^{−2}), and applying wavelet analysis to estimate $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ does not improve the agreement
between the two measurement systems. The removal of vehicle-induced
turbulence from the car measurements (and not the tripod) further decreases
$\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$, in turn increasing the bias between the car and tripod. Like $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$, the sonic heat flux ($\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$) measured by the mobile car in this study (shown in Fig. 8b) also has a low bias of 30 % to 50 %
compared to the tripod (MBE_{EC} = −0.05 K m s^{−1}). There is no improvement in the statistical measures if wavelet
analysis is used to estimate $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$. Despite a low bias noted in
$\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$, there is no low bias found in the sonic temperature variance
($\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$) measured on the instrumented car compared to the tripod
(shown in Fig. 9), where the MBE_{EC}=0.05 K^{2}.
Since the sonic anemometer is placed over the front bumper, which holds the
vehicle engine, there may potentially be some impact from its heat in our
measurements. While the effect of engine heat is probably more important in
cold ambient temperatures, there may still be an impact on the sonic
temperature (*T*) measured on the car in this study while driving, which
would likely result in $\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$ being biased high compared to an
instrumented car without engine heat effects.

The discrepancy between the car and the tripod for $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$and
$\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$ may be related to a mismatch in the flux footprint or possibly to the rapid flow distortion experienced at the location of the
sonic anemometer on the vehicle. The road produces a distinct upward heat
flux and an increase in $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ on sunny days, because it has a
significantly lower albedo than the surrounding grasses and farmland. On 22 August, we parked on the upwind side of the highway for approximately 30 min,
but the car was also parked on the downwind side of the highway during
assembly and disassembly of the tripod. For three independent 8 min periods,
the average $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$, $\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$, and $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$ on the upwind side of the highway are measured at 0.15 m^{2} s^{−2}, 0.46 K^{2}, and 0.085 K m s^{−1}, respectively. Downwind of the highway,
$\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$, $\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$, and $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$ are found to be larger, near 0.33 m^{2} s^{−2}, 0.68 K^{2}, and 0.109 K m s^{−1} on average
(from five independent samples), which are more consistent with measurements
made on the tripod, except for $\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$. The car-measured
$\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$ on the downwind side of the highway has a large standard
deviation (0.41 K^{2}) and a single outlier that skews the average.
Removing this outlier (where $\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$ = 1.39 K^{2}) reduces the average car-measured $\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$ downwind of the highway to 0.50 K^{2}, which is more consistent with the tripod; the 8 min sample with the anomalously large $\stackrel{\mathrm{\u203e}}{{{T}^{\prime}}^{\mathrm{2}}}$ does not have an anomalously large $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ or $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$. The findings in this study for
$\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ are similar to Gordon et al. (2012), who measured
$\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ = 0.27 m^{2} s^{−2} downwind of a four-lane highway on a sunny day.

To investigate the flux footprint of the tripod versus the instrumented car,
the footprint model of Kljun et al. (2015) is applied with $\stackrel{\mathrm{\u203e}}{u}$ = 2.5 m s^{−1}, a boundary layer height of *h* = 1500 m, a friction velocity
of *u*_{∗} = 0.35 m s^{−1}, an Obukhov length of *L* = −30 m, $\stackrel{\mathrm{\u203e}}{{{v}^{\prime}}^{\mathrm{2}}}$ = 1.5 m^{2} s^{−2}, and a wind direction that is assumed to be
perpendicular to the highway. These meteorological values represent
estimations based on measurements made on 22 August. For the car, *z*_{m} ≈ 1.7 m, but for the tripod, *z*_{m} ≈ 1.4 m. However, flow distortion
on the mobile car results in the measurements being representative of a
lower height than the height at which the instrumentation is installed.
Achberger and Bärring (1999) explored the displacement due to flow
distortion on a mini-bus and estimated that the displacement at 2 m height
was typically on the order of 0.2 m. Therefore, measurements obtained at
*z*_{m} = 1.7 m on the mobile car in this study are probably
representative of a slightly lower height between 1.5 and 1.6 m. For the
upper height limit of *z*_{m} = 1.7 m, the footprint model predicts that
the maximum location of influence to the flux is about 4.2 m upwind of the
measurement location. For *z*_{m} = 1.5 m, it is about 3.7 m upwind. Since
the tripod is positioned in the shoulder of the highway, 3.7 m upwind of the
tripod is near the center of the highway. Assuming the instrumented car is
in the lane closest to the tripod (or about 1.75 m from the edge of the
highway), the maximum location of influence to the flux is near 6 m or near
the edge of the highway furthest from the tripod. Therefore, when the car is
in the lane closest to the tripod, the measurements have a flux footprint
that includes less influence from the highway. The footprint model predicts
that the influence from the road is minimized when the car is driving in the
lane furthest from the tripod, but for measurements made during this study,
there is not a significant statistical difference in $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ and
$\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$ for the close versus far highway lane. The footprint model
applied here is strongly impacted by the mean wind speed $\stackrel{\mathrm{\u203e}}{u}$ – a
lower $\stackrel{\mathrm{\u203e}}{u}$ gives a location of maximum influence to the flux that is closer to the measurement system.

Another factor that may influence the velocity measurements made by the sonic anemometer is rapid distortion of the flow caused by the moving vehicle. Wyngard (1988) shows that the variance of scalar quantities (such as the sonic temperature or a gas concentration) remains unchanged during rapid flow distortion. The velocity variances, however, may be altered during stretching and compression of the flow, as it is forced to rise over the front end of the vehicle, similar to isotropic turbulence and flow over a symmetric hill (Britter et al., 1981; Gong and Ibbetson, 1989). If it is assumed that the low bias in the measured $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ on the car is caused by rapid flow distortion alone (i.e., no effect from the highway asphalt), then rapid distortion theory would predict a proportional increase in the velocity variance measured parallel to the vehicle motion. However, in the case of the measurements of $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ made during this study, there is likely a contribution from the rapid distortion of the flow in addition to a contribution from the flux footprint mismatch between the car and tripod, but it is not possible to separate the effects in this work.

For EC, there is no significant bias for $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ measured on the car compared to $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ measured on the tripod, as shown in Fig. 8a. The tripod measurements of $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ generally fall within the 95 % confidence interval of $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ measured on the car (see Sect. 3.4.2). However, there are instances where $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ measured by the two systems differ significantly, and this suggests that a better estimate of $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ can probably be obtained by averaging multiple passes. The horizontal momentum flux, $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{v}^{\prime}}$, measured on the tripod does not agree with measurements made on the mobile car (not shown), and when sampling errors are considered, $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{v}^{\prime}}$ measured on the car is not found to be statistically different than 0 within the 66 % confidence interval.

## 3.3 Velocity spectra

Figure 10 displays the binned power spectral density (multiplied by
frequency) of the velocity components for measurement Pass 5 (Fig. 10a),
Pass 7 (Fig. 10b), and Pass 8 (Fig. 10c) from Track #1. These three
measurement passes have been chosen, since they demonstrate unique features
in the car spectra, which are representative of the spectra from the
remaining measurement passes not shown (see Fig. S6). The frequencies are
normalized to give a wavelength as $\mathit{\lambda}=\stackrel{\mathrm{\u203e}}{u}/f$, where *f* is the
frequency (Hz) and $\stackrel{\mathrm{\u203e}}{u}$ is the mean ambient wind on the tripod or the car relative flow on the mobile car. Each panel displays the spectra of *u* (top), *v* (middle), and *w* (bottom). In general, the shape of the spectra
measured on the mobile car agree well with the spectra measured by the
tripod; however, there are some notable differences: (1) unlike the tripod,
the power spectra of *u* and *v* measured on the car during Pass 7 and 8
increase at high frequencies (*λ*<5 m). This increase may be related
to white noise in the measured signal or perhaps to aliasing and is present in
about 75 % of the measured spectra from Track #1. Langford et al. (2015) show that the power spectra of the sonic temperature increase
linearly with a +1 at high frequencies (in the inertial subrange) in the
presence of white noise, resembling the findings in this study for *u* and
*v*. One potential source of white noise in the measured horizontal velocity
components may be road unevenness (Schiehlen, 2006). Belušić et al. (2014)
found distinct peaks near a frequency of 7 Hz in their car-measured *v* spectra, which they attribute to frame vibrations, and by comparing the sonic
measurements to GPS–INS motion, they concluded that road unevenness did not
impact the high frequency portion of the velocity spectra. (2) For *u* in
Pass 7 and 8, as *λ* increases past 100 m, the power spectral density
increases on the car, while on the tripod, the power spectral density
decreases. (3) In Pass 7 and 8, *w* appears to be under-sampled, since the
car spectra do not extend through the entire inertial subrange. Therefore,
sampling at high vehicle speeds (> 15 m s^{−1}) would probably
benefit from a sampling rate greater than 40 Hz. Additionally, in Pass 5 and
8, there is a general underestimation of the power spectral density of *w* on
the car compared to the tripod for *λ* between about 5 to 80 m, and
this underestimation is a common feature in the measured car spectra.

## 3.4 Measurement uncertainties

### 3.4.1 Flow distortion correction angle, *θ*

Despite the rather strong relationship between the measured vertical
velocity (*W*) and the measured longitudinal velocity (*U*) discussed in
Sect. 2.4, there is still an uncertainty in the rotation angle (*θ*)
used to correct for the effect of flow distortion on the vertical velocity.
The median of *θ* calculated using all binned values is
7.54^{∘}, with the lower and upper quartile (25th and
75th) being 7.38 and 7.70^{∘}, respectively (IQR = 0.32^{∘}). If *θ* = *Q*25 = 7.38^{∘} is used for the flow distortion correction instead, the mean
vertical velocity measured on the car during all measurement passes
increases, giving $\stackrel{\mathrm{\u203e}}{w}$ = 0.06 m s^{−1} (using *θ* = *Q*50 = 7.54^{∘} gives $\stackrel{\mathrm{\u203e}}{w}$ = 0.00 m s^{−1}). In addition,
there is an increase in the magnitude of $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$, $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$, and $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$, giving a marginally better statistical agreement between the car and tripod for $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$, as shown in Table 5. These
results demonstrate that reducing *θ* to give $\stackrel{\mathrm{\u203e}}{w}$ > 0 m s^{−1} is not sufficient to improve the agreement among all turbulence statistics and will not remove the bias noted in Sect. 3.2 for $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$. Similarly, increasing *θ* from 7.54^{∘} does not remove the bias or improve the agreement between
the car and tripod.

### 3.4.2 Sampling errors

A significant concern when obtaining atmospheric measurements from an instrumented mobile car is the impact of sampling errors. Sampling errors on the mobile car may result from (i) the use of a record length that is too short to be representative of an ensemble mean, (ii) non-stationarity of the flow introduced by microscale variations or inhomogeneities in the terrain and surrounding structures (i.e., trees, buildings), or (iii) white noise and persistent structured signals introduced by vehicle resonance and vibrations.

In this work, three methods to quantity the random measurement uncertainty
are investigated: (1) the method of Finkelstein and Sims (2001), referred to
as F&S (Eq. 17, denoted as *δ*_{FS}); (2) the method of Mann and Lenschow (1994), referred to as M&L (Eq. 15, denoted as *δ*_{ML}); and (3) the method of Lenschow et al. (2000) (Eq. 22). F&S and M&L
give an estimate of the overall random measurement uncertainty, while
Lenschow et al. (2000) gives an estimate of the random measurement uncertainty
attributed only to white noise in the measured signal. The method of
Lenschow et al. (2000) does not include contributions from persistent
structured signals that may occur at a specific frequency (i.e., from
vehicle resonance or some other cause of vibrations, such as speed bumps).
*δ*_{FS} and *δ*_{ML} give 1 standard deviation of the random
measurement uncertainty of a measured variance or covariance for the
averaging period *T*_{m}, which is demonstrated by Rannik et al. (2009) to be nearly
equivalent to the standard error of the variance or covariance. Thus, in
this work, we define the 68 % confidence interval as the range *F*±*δ* and likewise the 95 % confidence interval as the range *F*±1.96*δ*, where *F* is the measured variance or covariance. When the
confidence interval of a variance or covariance includes the value measured
on the tripod, then measurements are deemed consistent between the two
systems in that confidence interval (for that measurement pass). Figures 11
to 13 display the random measurement uncertainty of the measured variances
and covariances, calculated using these three methodologies.

The random uncertainty estimates calculated from M&L and F&S agree
well on the mobile car platform for velocity variances when *m* = 30 s.
However, for *m* = 30 s, F&S tends to give a slightly greater magnitude of random measurement uncertainty than M&L for covariances (i.e., Fig. 13). This is similar to the findings of Finkelstein and Sims (2001), who
note that the method of F&S contains a contribution from both the
autocovariance and cross-covariance function, leading to a larger magnitude
and more conservative estimate of the sampling error compared to M&L.
Rannik et al. (2016) note that F&S gives an estimate of the “total”
random measurement uncertainty

The random measurement uncertainty calculated from F&S and M&L scales
approximately linearly with increasing magnitude of the velocity variance or
covariance, as shown in Figs. 11 to 13. For Track #2, there are
several instances where $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ is large (i.e., 2 to 5 m^{2} s^{−2}) and *δ*_{FS} is on the order of $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ . Thus, $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ measured on Track #2 is not statistically different than 0
in the 95 % confidence interval for some measurement passes. A trend in
the velocity record results in an autocorrelation function that does not
fall to 0 as expected and instead remains elevated at large time lags.
This suggests that *δ*_{FS} in this study includes a contribution from non-stationarity in the record, which is consistent with the conclusions for measurements made on stationary towers from Rannik et al. (2016), who found that *δ*_{FS} continues to increase as *m* is increased to 300 s.

Reconstructing the time series using wavelet analysis produces a filtered
time series, where the resolved low frequency contributions are excluded.
Applying F&S to the reconstructed time series gives an estimate of
*δ*_{FS} for the wavelet variances and covariances (shown in Figs. 11 to 13 as diamonds). For $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$, wavelet estimates of *δ*_{FS}
follow a similar trend to the uncertainty estimates found using the
unfiltered time series – that is, as the magnitude of the wavelet variance
increases, so does *δ*_{FS}. However, for times when wavelet analysis predicts a smaller $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$, *δ*_{FS} is also found to be
proportionally reduced.

For the measurement tracks investigated here, the use of a linear fit to
estimate *δ*_{L} gives a much larger uncertainty than Eq. (21), as
shown in Figs. 11 and 12. In the case of the vertical velocity, *δ*_{L} estimated using a linear fit extrapolation is 3 to 4 times larger than the total random measurement uncertainty according to *δ*_{FS}. *δ*_{L} is expected to represent a contribution to the total random measurement uncertainty, and therefore *δ*_{L}<*δ*_{FS} (Rannik et
al., 2016). This suggests that the linear fit significantly underestimates
the true variance and overestimates the amount of white noise for *w*. If a
power law fit (Eq. 21) is used instead of a linear fit, *δ*_{L} is reduced, and for several measurements passes *δ*_{L}<*δ*_{FS}. The difficulty of estimating *δ*_{L} for *w* on the car is not unexpected,
since *w* has an integral timescale (ITS) of 0.05 to 0.1 s for vehicle
speeds near 20 m s^{−1}, and this is only 2 to 4 times the sampling
interval of the sonic anemometer. This limits the amount of autocovariance
function time lags that lie within the inertial subrange, giving a poor fit.
Lenschow et al. (2000) note that, for a successful power law fit to the
autocovariance function, the ITS must be “several times larger” than the
sampling interval of the instrument. For *w* measured on the tripod, the use
of Eq. (21) gives undefined $\stackrel{\mathrm{\u203e}}{{{\mathit{\u03f5}}^{\prime}}^{\mathrm{2}}}$, while a linear fit gives *δ*_{L}>*δ*_{FS}, as shown in Fig. 12b.

Compared to *w*, the measured horizontal velocity components on the car
(*u*, *v*) have a larger ITS (on the order of 1 s) and a larger
signal-to-noise ratio (SNR). Rannik et al. (2016) argue that the method
proposed by Lenschow et al. (2000) is best suited for closed-path sensors
as opposed to open-path sensors and high-precision instrumentation such as
sonic anemometers. They found that the method of Lenschow et al. (2000)
gives a relatively unbiased estimate of the white noise when the SNR is
small and applied the method to estimate *δ*_{L} only for *w* (not for *u* or *v*). For *u* and *v* in this study, *δ*_{L} typically
represents a small contribution to the total random measurement uncertainty,
except for weaker signals (i.e., lower measured horizontal variances). The
presence of white noise in the measured *u* and *v* signals is also
supported by the spectra shown in Fig. 10b, where a near +1 slope appears
at high frequencies within the inertial subrange. This is not the case for
*w*, where the spectra do not show a +1 slope at high frequencies;
hence, *w* spectra have no evidence of white noise impacting the measured
signal. This may suggest that *δ*_{L} overestimates the magnitude of white noise present in *w*, and so *δ*_{L} is likely not a reliable
estimate of white noise in the vertical velocity for car measurements
made at high vehicle speeds near 20 m s^{−1}.

In addition to Track #1 and Track #2, the car was driven on a gravel
road at relatively high vehicle speeds (*s* between 20 and 23 m s^{−1})
for a short (< 5 min) period. The effect of the gravel road is
investigated by splitting the short period into non-overlapping intervals
of 49 s (yielding 5 unique samples) and performing the same analysis as
outlined in Sect. 2. The car measurements on the gravel road are similar to
car measurements obtained on the paved road for a comparable *s*. The
magnitude of the variances and covariances on the gravel road are consistent
with those measured on the paved road within the 95 % confidence
interval, and the uncertainty estimates (*δ*_{FS}, *δ*_{ML}, and
*δ*_{L}) are the same order of magnitude. The measured velocity
variances and uncertainty analysis for the gravel road are displayed in the
Supplement (Fig. S7). These measurements suggest that the road
surface types investigated in this study have a limited influence on the
measured turbulence statistics.

### 3.4.3 Tripod velocity record contamination from passing traffic

Since the study was designed to investigate measurements in non-idealized conditions, the highway locations have public access; therefore, other vehicle traffic was present during the measurements. The traffic consisted largely of passenger vehicles (such as cars, pickup trucks, sport utility vehicles, and minivans), but the traffic on 22 August was more significant and was comprised of occasional large trucks (dump trucks and tractor-trailers). For measurement passes on 22 August (with video recordings available on the tripod), the dashboard camera recorded between 26 and 40 total passing vehicles, of which 0 to 4 were large trucks. The car takes about 45 s to complete a track, but on the tripod, the equivalent averaging period is between 6 to 8 min. For some measurement passes, the mobile car does not experience any traffic contamination, but this is not the case for the tripod. Therefore, the tripod will measure a different composition and amount of passing traffic than the car, potentially leading to differences in the measurements made by the two systems.

Large trucks produce a significant amount of vehicle-induced turbulence, but passenger cars and sport utility vehicles produce much less in comparison (Miller et al., 2019; Gordon et al., 2012). Furthermore, the wake has limited lateral spread relative to the vehicle travel direction (Kim et al., 2016), except perhaps for times with significant advection, so the most noticeable effect on the tripod will be from traffic in the adjacent highway lane (i.e., closet to the tripod). For measurements on the car, passing traffic (particularly large trucks) is found to enhance the measured velocity variances (i.e., Fig. 7c). Like the car, the main effect of passing traffic on the tripod measurements would also be an enhancement of the velocity variances. Thus, for times when there is no traffic contamination on the car, the differences shown in Fig. 6 between the car and tripod-measured velocity variances may be underestimated, since the tripod velocity variances are enhanced due to passing traffic but the car measurements are not. Therefore, the presence of traffic measured by the tripod and not the car introduces an additional uncertainty into the measurement comparisons shown in Sect. 3.

The results presented in Sect. 3 demonstrate that the instrumented car design used in this study can successfully measure the mean atmospheric boundary layer close to the surface, but the car measurements may vary significantly based on the surrounding features such as trees, buildings, and other traffic. Therefore, the interpretation of the car-based measurements depends largely on the specific application, since the car may measure turbulence that is localized and not represented in single-point measurements made at a stationary tower. In the previous study of Belušić et al. (2014), there was limited upwind surface obstructions and no other traffic during their measurements. Despite the more idealized environment, their measurements revealed times when the horizontal velocity variances ($\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{{v}^{\prime}}^{\mathrm{2}}}$) measured on the car were significantly larger than a nearby stationary tower, and they suggest that intense, temporally limited flow structures are to blame. These events dominate the measurements made on the car but not on the tripod, since the averaging period is longer. In this investigation, $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{{v}^{\prime}}^{\mathrm{2}}}$ on the car calculated using EC are also found to be much larger than measured on the tripod for some measurement passes (i.e., a factor between 2 and 5 for $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$, with ${\mathrm{RMSE}}_{\mathrm{EC}\phantom{\rule{0.125em}{0ex}}\mathrm{Car}}/{\mathrm{Mean}}_{\mathrm{EC}\phantom{\rule{0.125em}{0ex}}\mathrm{Tripod}}={\mathrm{NRMSE}}_{\mathrm{EC}}$ ≈ 114 %). When the measurement uncertainty in Sect. 3 is considered, these large $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ are not statistically different than 0 in the 95 % confidence interval, since ${\mathit{\delta}}_{\mathrm{FS}}\approx \stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$. Applying wavelet analysis to calculate $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{{v}^{\prime}}^{\mathrm{2}}}$ gives significantly reduced magnitudes for some measurement passes, particularly those measurement passes with the largest estimated EC variances. This results in an improved agreement between the mobile car and tripod for $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{{v}^{\prime}}^{\mathrm{2}}}$ (for $\stackrel{\mathrm{\u203e}}{{{u}^{\prime}}^{\mathrm{2}}}{\mathrm{RMSE}}_{\mathrm{W}\phantom{\rule{0.125em}{0ex}}\mathrm{Car}}/{\mathrm{Mean}}_{\mathrm{EC}\phantom{\rule{0.125em}{0ex}}\mathrm{Tripod}}={\mathrm{NRMSE}}_{\mathrm{W}}$ ≈ 60 %). The improved agreement using wavelet analysis suggests that wavelet analysis resolves length scales near and exceeding the length of the measurement track (i.e., 1000 m); in this study, the change in surface features on Track #2 (from a windbreak to an open field) may yield an artificial low frequency contribution in the velocity record. Thus, when measuring from an instrumented car, it is important to be aware of changes in terrain and land usage, which can strongly impact the near-ground measurements.

Evidence from this investigation shows that passing traffic (especially large trucks) can also lead to an increase in the velocity variances measured on the car. However, if the passing traffic is sporadic, the resulting increase in the measured velocity variances from vehicle-induced turbulence can be identified and removed using wavelet analysis. In this study, for a measurement pass that experienced a passing heavy-duty truck and sport utility vehicle, removing the times when the traffic passes the mobile car (9 out of 46 s) decreases the turbulent kinetic energy by about 10 %. This highlights the importance of video recordings in conjunction with sonic anemometer measurements on a car, so that times with possible traffic contamination can be identified in applications where its measurement is not intended.

The sampling uncertainties in Sect. 3 suggest that it is possible to measure
a statistically significant vertical momentum flux on the mobile car at
vehicle speeds near 20 m s^{−1}. $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ measured on the car is
typically found to be consistent with the tripod within the 95 % confidence
interval, but for some passes, $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ measured on the car is
small (< 0.06 m^{2} s^{−2}) and not statistically different than
0 in the 95 % confidence interval. Therefore, for measurements obtained
on the mobile car, a better estimate of $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ can probably be
obtained by averaging multiple passes with a spatial extent of 10s of
kilometers. Random measurement uncertainty estimates of $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$
by F&S and M&L (which give 1 standard deviation of the uncertainty)
have magnitudes that are typically 10 % to 40 % of the measured flux.
Furthermore, there is no significant bias in $\stackrel{\mathrm{\u203e}}{{u}^{\prime}{w}^{\prime}}$ measured on
the car when the entire set of measurement passes is considered
(${\mathrm{MBE}}_{\mathrm{EC}\phantom{\rule{0.125em}{0ex}}\mathrm{Car}}/{\mathrm{Mean}}_{\mathrm{EC}\phantom{\rule{0.125em}{0ex}}\mathrm{Tripod}}={\mathrm{NMBE}}_{\mathrm{EC}}$ ≈ −4 % and ${\mathrm{MBE}}_{\mathrm{W}\phantom{\rule{0.125em}{0ex}}\mathrm{Car}}/{\mathrm{Mean}}_{\mathrm{EC}\phantom{\rule{0.125em}{0ex}}\mathrm{Tripod}}={\mathrm{NMBE}}_{\mathrm{W}}$ ≈ −14 %).

The vertical velocity ($\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$) and vertical sonic heat flux
($\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$) measured in this study are found to be biased low compared to measurements made on the tripod (NMBE_{EC} ≈ −38 % for both $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ and $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$). The low bias on the car is
probably due to the combination of two factors: (1) the footprint measured
by the car contains less of the low-albedo highway than the tripod, and (2) rapid flow distortion at the measurement location on the car. Interestingly, there is evidence of a similar low bias in $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ (but not $\stackrel{\mathrm{\u203e}}{{w}^{\prime}{T}^{\prime}}$) measured by the car in Belušić et al. (2014), where only 4 out of the 19 completed passes measured a greater $\stackrel{\mathrm{\u203e}}{{{w}^{\prime}}^{\mathrm{2}}}$ on the car than
the stationary tower (i.e., their Fig. 4). This demonstrates that wind
tunnel testing or computational flow modeling of each specific instrumented
car design may be useful to quantify the effects of rapid flow distortion on
the measured velocity variances and covariances. Applying the method of
Lenschow et al. (2000) to estimate the magnitude of white noise in the
measured vertical velocity signal at vehicle speeds near 20 m s^{−1}
likely underestimates the true signal variance and overestimates the amount
of white noise and therefore is not recommended.

The mean wind speed and mean wind direction were found to be consistent with
measurements made on the tripod. For $\stackrel{\mathrm{\u203e}}{u}$ measured on Track #1 and Track #2, the NMBE ≈ 2 % and NRMSE ≈ 22 %, respectively. Even a short averaging period of 10 s for
car measurements made at a vehicle speed near 20 m s^{−1} provides a
reliable estimate of mean wind direction on the car; for about 250 unique
intervals on 20 and 22 August, the interquartile range of the wind direction is 42 and 34^{∘}, respectively. Despite the rather consistent wind
direction, the mean wind speed in any individual 10 s averaging period may
vary considerably; the interquartile range for $\stackrel{\mathrm{\u203e}}{u}$ is 1.3 and 1.9 m s^{−1} on 20 and 22 August, respectively. The large variation in the 10 s mean
wind speed likely represents more localized flow that exists in a specific
location. Therefore, the instrumented car may prove invaluable for studies
that require precise measurement of localized flow, providing simultaneous
measurements of wind speed and direction over a large domain. This study
shows that, even when the sonic anemometer is placed particularly close to
the vehicle (compared with Belušić et al., 2014, for example), it is still possible to correct for flow distortion effects and to obtain measurements of the mean wind and turbulence that are consistent (within the 95 %
confidence interval) of those measured by a nearby stationary tripod.

The results presented in this investigation demonstrate that car-based measurements of turbulence require care when selecting the appropriate spatial and temporal averaging and when selecting the measurement location to ensure that the measurements obtained are representative of the specific application. This is demonstrated in our measurements, where the highway surface or flux footprint, upwind obstructions, and passing traffic are all found to have a significant effect on the measured values but are not necessarily errors, since they do represent real features that can generate atmospheric turbulence.

The data used to generate the figures and complete the analysis presented herein are available online at https://doi.org/10.5683/SP3/IBBDTF (Gordon, 2022). The code to complete the wavelet analysis is available online at https://github.com/chris-torrence/wavelets (last access: 1 December 2021; Torrence and Compo, 1998b).

The supplement related to this article is available online at: https://doi.org/10.5194/amt-15-6563-2022-supplement.

SJM and MG designed and performed the experiment. SJM completed the analysis of the measured data and complied the results. SJM prepared the manuscript with contributions from Mark Gordon.

The contact author has declared that neither of the authors has any competing interests.

Publisher’s note: Copernicus Publications remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.

We thank Peter Taylor for providing the SUV used in this study and for his assistance during the experimental data collection (as a driver in the experiment) on 30 August.

This research has been supported by the Natural Sciences and Engineering Research Council of Canada (grant no. RGPIN 2015–04292).

This paper was edited by Cléo Quaresma Dias-Junior and reviewed by Luca Mortarini and one anonymous referee.

Achberger, C. and Bärring, L.: Correction of surface air temperature measurements from a mobile platform, Agr. Forest Meteorol., 98–99, 227–238, https://doi.org/10.1016/s0168-1923(99)00099-4, 1999.

Anderson, A. R., Chapman, M., Drobot, S. D., Tadesse, A., Lambi, B., Wiener, G., and Pisano, P.: Quality of mobile air temperature and atmospheric pressure observations from the 2010 Development Test Environment Experiment, J. Appl. Meteorol. Clim., 51, 691–701, https://doi.org/10.1175/jamc-d-11-0126.1, 2012.

Aristodemou, E., Boganegra, L. M., Mottet, L., Pavlidis, D., Constantinou, A., Pain, C., Robins, A., and ApSimon, H.: How tall buildings affect turbulent air flows and dispersion of pollution within a neighbourhood, Environ. Pollut., 233, 782–796, https://doi.org/10.1016/j.envpol.2017.10.041, 2018.

Armi, L. and Mayr, G. J.: Continuously stratified flows across an alpine crest with a pass: Shallow and Deep Föhn, Q. J. Roy. Meteor. Soc., 133, 459–477, https://doi.org/10.1002/qj.22, 2007.

Belušić, D., Lenschow, D. H., and Tapper, N. J.: Performance of a mobile car platform for mean wind and turbulence measurements, Atmos. Meas. Tech., 7, 1825–1837, https://doi.org/10.5194/amt-7-1825-2014, 2014.

Bogren, J. and Gustavsson, T.: Nocturnal Air and road surface temperature variations in complex terrain, Int. J. Climatol., 11, 443–455, https://doi.org/10.1002/joc.3370110408, 1991.

Bonin, T. A., Newman, J. F., Klein, P. M., Chilson, P. B., and Wharton, S.: Improvement of vertical velocity statistics measured by a Doppler lidar through comparison with sonic anemometer observations, Atmos. Meas. Tech., 9, 5833–5852, https://doi.org/10.5194/amt-9-5833-2016, 2016.

Britter, R. E., Hunt, J. C., and Richards, K. J.: Air flow over a two-dimensional hill: Studies of velocity speed-up, roughness effects and turbulence, Q. J. Roy. Meteor. Soc., 107, 91–110, https://doi.org/10.1002/qj.49710745106, 1981.

Conte, M., Contini, D., and Held, A.: Multiresolution decomposition and wavelet analysis of urban aerosol fluxes in Italy and Austria, Atmos. Res., 248, 105267, https://doi.org/10.1016/j.atmosres.2020.105267, 2021.

Curry, M., Hanesiak, J., Kehler, S., Sills, D. M., and Taylor, N. M.: Ground-based observations of the thermodynamic and kinematic properties of Lake-Breeze Fronts in southern Manitoba, Canada, Bound.-Lay. Meteorol., 163, 143–159, https://doi.org/10.1007/s10546-016-0214-1, 2017.

de Boer, G., Waugh, S., Erwin, A., Borenstein, S., Dixon, C., Shanti, W., Houston, A., and Argrow, B.: Measurements from mobile surface vehicles during the Lower Atmospheric Profiling Studies at Elevation – a Remotely-piloted Aircraft Team Experiment (LAPSE-RATE) , Earth Syst. Sci. Data, 13, 155–169, https://doi.org/10.5194/essd-13-155-2021, 2021.

Finkelstein, P. L. and Sims, P. F.: Sampling error in eddy correlation flux measurements, J. Geophys. Res.-Atmos., 106, 3503–3509, https://doi.org/10.1029/2000jd900731, 2001.

Göckede, M., Kittler, F., and Schaller, C.: Quantifying the impact of emission outbursts and non-stationary flow on eddy-covariance CH_{4} flux measurements using wavelet techniques, Biogeosciences, 16, 3113–3131, https://doi.org/10.5194/bg-16-3113-2019, 2019.

Gong, W. and Ibbetson, A.: A wind tunnel study of turbulent flow over Model Hills, Bound.-Lay. Meteorol., 49, 113–148, https://doi.org/10.1007/bf00116408, 1989.

Gordon, M.: The measurement of mean wind, variances and covariances from an instrumented mobile car in a rural environment, Borealis, V1 [data set], https://doi.org/10.5683/SP3/IBBDTF, 2022.

Gordon, M., Staebler, R. M., Liggio, J., Makar, P., Li, S.-M., Wentzell, J., Lu, G., Lee, P., and Brook, J. R.: Measurements of enhanced turbulent mixing near Highways, J. Appl. Meteorol. Clim., 51, 1618–1632, https://doi.org/10.1175/jamc-d-11-0190.1, 2012.

Gromke, C. and Blocken, B.: Influence of avenue-trees on air quality at the Urban Neighborhood Scale. part I: Quality assurance studies and turbulent schmidt number analysis for RANS CFD Simulations, Environ. Pollut., 196, 214–223, https://doi.org/10.1016/j.envpol.2014.10.016, 2015.

Hanlon, T. and Risk, D.: Using computational fluid dynamics and field experiments to improve vehicle-based wind measurements for environmental monitoring, Atmos. Meas. Tech., 13, 191–203, https://doi.org/10.5194/amt-13-191-2020, 2020.

Hertwig, D., Gough, H. L., Grimmond, S., Barlow, J. F., Kent, C. W., Lin, W. E., Robins, A. G., and Hayden, P.: Wake characteristics of tall buildings in a realistic urban canopy, Bound.-Lay. Meteorol., 172, 239–270, https://doi.org/10.1007/s10546-019-00450-7, 2019.

Hunt, J. C. R., Poulton, E. C., and Mumford, J. C.: The effects of wind on people; new criteria based on wind tunnel experiments, Build. Environ., 11, 15–28, https://doi.org/10.1016/0360-1323(76)90015-9, 1976.

Kim, Y., Huang, L., Gong, S., and Jia, Q. C.: A new approach to quantifying vehicle induced turbulence for complex traffic scenarios, Chinese J. Chem. Eng., 1, 71–78, https://doi.org/10.1016/j.cjche.2015.11.025, 2016.

Kljun, N., Calanca, P., Rotach, M. W., and Schmid, H. P.: A simple two-dimensional parameterisation for Flux Footprint Prediction (FFP), Geosci. Model Dev., 8, 3695–3713, https://doi.org/10.5194/gmd-8-3695-2015, 2015.

Krayenhoff, E. S., Jiang, T., Christen, A., Martilli, A., Oke, T. R., Bailey, B. N., Nazarian, N., Voogt, J. A., Giometto, M. G., Stastny, A., and Crawford, B. R.: A multi-layer urban canopy meteorological model with trees (BEP-tree): Street tree impacts on pedestrian-level climate, Urban Climate, 32, 100590, https://doi.org/10.1016/j.uclim.2020.100590, 2020.

Langford, B., Acton, W., Ammann, C., Valach, A., and Nemitz, E.: Eddy-covariance data with low signal-to-noise ratio: time-lag determination, uncertainties and limit of detection, Atmos. Meas. Tech., 8, 4197–4213, https://doi.org/10.5194/amt-8-4197-2015, 2015.

Lee, J. P. and Lee, S. J.: PIV analysis on the shelter effect of a bank of real fir trees, J. Wind Eng. Ind. Aerod., 110, 40–49, https://doi.org/10.1016/j.jweia.2012.07.003, 2012.

Lenschow, D. H., Mann, J., and Kristensen, L.: How long is long enough when measuring fluxes and other turbulence statistics?, J. Atmos. Ocean. Tech., 11, 661–673, https://doi.org/10.1175/1520-0426(1994)011<0661:HLILEW>2.0.CO;2, 1994.

Lenschow, D. H., Wulfmeyer, V., and Senff, C.: Measuring second- through fourth-order moments in Noisy Data, J. Atmos. Ocean. Tech., 17, 1330–1347, https://doi.org/10.1175/1520-0426(2000)017<1330:MSTFOM>2.0.CO;2, 2000.

Lyu, J., Wang, C. M., and Mason, M. S.: Review of models for predicting wind characteristics behind windbreaks, J. Wind Eng. Ind. Aerod., 199, 104117, https://doi.org/10.1016/j.jweia.2020.104117, 2020.

Mahrt, L., Richardson, S., Seaman, N., and Stauffer, D.: Turbulence in the nocturnal boundary layer with light and variable winds, Q. J. Roy. Meteor. Soc., 138, 1430–1439, https://doi.org/10.1002/qj.1884, 2012.

Mann, J. and Lenschow, D. H.: Errors in airborne flux measurements, J. Geophys. Res., 99, 14519, https://doi.org/10.1029/94jd00737, 1994.

Markowski, P. M., Lis, N. T., Turner, D. D., Lee, T. R., and Buban, M. S.: Observations of near-surface vertical wind profiles and vertical momentum fluxes from vortex-SE 2017: Comparisons to Monin-Obukhov similarity theory, Mon. Weather Rev., 147, 3811–3824, https://doi.org/10.1175/mwr-d-19-0091.1, 2019.

Mauder, M., Cuntz, M., Drüe, C., Graf, A., Rebmann, C., Schmid, H. P., Schmidt, M., and Steinbrecher, R.: A strategy for quality and uncertainty assessment of long-term eddy-covariance measurements, Agr. Forest Meteorol., 169, 122–135, https://doi.org/10.1016/j.agrformet.2012.09.006, 2013.

Mayr, G. J. and Armi, L.: Föhn as a response to changing upstream and downstream air masses, Q. J. Roy. Meteor. Soc., 134, 1357–1369, https://doi.org/10.1002/qj.295, 2008.

Miller, S. J., Gordon, M., Staebler, R. M., and Taylor, P. A.: A study of the spatial variation of vehicle-induced turbulence on highways using measurements from a mobile platform, Bound.-Lay. Meteorol., 171, 1–29, https://doi.org/10.1007/s10546-018-0416-9, 2019.

Mochida, A., Tabata, Y., Iwata, T., and Yoshino, H.: Examining tree canopy models for CFD prediction of wind environment at pedestrian level, J. Wind Eng. Ind. Aerod., 96, 1667–1677, https://doi.org/10.1016/j.jweia.2008.02.055, 2008.

Paterna, E., Crivelli, P., and Lehning, M.: Decoupling of mass flux and turbulent wind fluctuations in drifting snow, Geophys. Res. Lett., 43, 4441–4447, https://doi.org/10.1002/2016gl068171, 2016.

Rannik, Ü., Mammarella, I., Aalto, P., Keronen, P., Vesala, T., and Kulmala, M.: Long-term aerosol particle flux observations part I: Uncertainties and time-average statistics, Atmos. Environ., 43, 3431–3439, https://doi.org/10.1016/j.atmosenv.2009.02.049, 2009.

Rannik, Ü., Peltola, O., and Mammarella, I.: Random uncertainties of flux measurements by the eddy covariance technique, Atmos. Meas. Tech., 9, 5163–5181, https://doi.org/10.5194/amt-9-5163-2016, 2016.

Salmond, J. A.: Wavelet analysis of intermittent turbulence in a very stable nocturnal boundary layer: Implications for the vertical mixing of ozone, Bound.-Lay. Meteorol., 114, 463–488, https://doi.org/10.1007/s10546-004-2422-3, 2005.

Salmond, J. A. and McKendry, I. G.: A review of turbulence in the very stable nocturnal boundary layer and its implications for Air Quality, Prog. Phys. Geog., 29, 171–188, https://doi.org/10.1191/0309133305pp442ra, 2005.

Schaller, C., Göckede, M., and Foken, T.: Flux calculation of short turbulent events – comparison of three methods, Atmos. Meas. Tech., 10, 869–880, https://doi.org/10.5194/amt-10-869-2017, 2017.

Schiehlen, W.: White noise excitation of road vehicle structures, Sadhana, 31, 487–503, https://doi.org/10.1007/bf02716788, 2006.

Smith, S. A., Brown, A. R., Vosper, S. B., Murkin, P. A., and Veal, A. T.: Observations and simulations of cold air pooling in Valleys, Bound.-Lay. Meteorol., 134, 85–108, https://doi.org/10.1007/s10546-009-9436-9, 2010.

Starkenburg, D., Metzger, S., Fochesatto, G. J., Alfieri, J. G., Gens, R., Prakash, A., and Cristóbal, J.: Assessment of despiking methods for turbulence data in Micrometeorology, J. Atmos. Ocean. Tech., 33, 2001–2013, https://doi.org/10.1175/jtech-d-15-0154.1, 2016.

Straka, J. M., Rasmussen, E. N., and Fredrickson, S. E.: A mobile mesonet for Finescale Meteorological Observations, J. Atmos. Ocean. Tech., 13, 921–936, https://doi.org/10.1175/1520-0426(1996)013<0921:AMMFFM>2.0.CO;2, 1996.

Strunin, M. A. and Hiyama, T.: Applying wavelet transforms to analyse aircraft-measured turbulence and turbulent fluxes in the atmospheric boundary layer over Eastern Siberia, Hydrol. Process., 18, 3081–3098, https://doi.org/10.1002/hyp.5750, 2004.

Su, J., Wang, L., Gu, Z., Song, M., and Cao, Z.: Effects of real trees and their structure on pollutant dispersion and flow field in an idealized Street Canyon, Atmos. Pollut. Res., 10, 1699–1710, https://doi.org/10.1016/j.apr.2019.07.001, 2019.

Taylor, N. M., Sills, D. M., Hanesiak, J. M., Milbrandt, J. A., Smith, C. D., Strong, G. S., Skone, S. H., McCarthy, P. J., and Brimelow, J. C.: The understanding severe thunderstorms and Alberta Boundary Layers Experiment (unstable) 2008, B. Am. Meteorol. Soc., 92, 739–763, https://doi.org/10.1175/2011bams2994.1, 2011.

Taylor, P. A. and Salmon, J. R.: A model for the correction of surface wind data for sheltering by upwind obstacles, J. Appl. Meteorol., 32, 1683–1694, https://doi.org/10.1175/1520-0450(1993)032<1683:AMFTCO>2.0.CO;2, 1993.

Torrence, C. and Compo, G. P.: A practical guide to wavelet analysis, B. Am. Meteorol. Soc., 79, 61–78, https://doi.org/10.1175/1520-0477(1998)079<0061:APGTWA>2.0.CO;2, 1998a.

Torrence, C. and Compo, G. P.: A practical guide to wavelet analysis, GitHub [code], https://github.com/chris-torrence/wavelets (last access: 1 December 2021), 1998b.

Turner, D. B.: Comparison of three methods for calculating the standard deviation of the wind direction, J. Clim. Appl. Meteorol., 25, 703–707, https://doi.org/10.1175/1520-0450(1986)025<0703:COTMFC>2.0.CO;2, 1986.

Van de Wiel, B. J., Moene, A. F., Jonker, H. J., Baas, P., Basu, S., Donda, J. M., Sun, J., and Holtslag, A. A.: The minimum wind speed for sustainable turbulence in the nocturnal boundary layer, J. Atmos. Sci., 69, 3116–3127, https://doi.org/10.1175/jas-d-12-0107.1, 2012.

von der Heyden, L., Deventer, M. J., Graus, M., Karl, T., Lamprecht, C., and Held, A.: Aerosol particles during the Innsbruck Air Quality Study (INNAQS): The impact of transient fluxes on total aerosol number exchange, Atmos. Environ., 190, 389–400, https://doi.org/10.1016/j.atmosenv.2018.07.041, 2018.

White, L. D.: Mobile observations of a quasi-frontal transient moisture boundary in the Deep South, Weather Forecast., 29, 1356–1373, https://doi.org/10.1175/waf-d-14-00009.1, 2014.

Wilczak, J. M., Oncley, S. P., and Stage, S. A.: Sonic anemometer tilt correction algorithms, Bound.-Lay. Meteorol., 99, 127–150, https://doi.org/10.1023/a:1018966204465, 2001.

Wulfmeyer, V., Pal, S., Turner, D. D., and Wagner, E.: Can water vapour raman lidar resolve profiles of turbulent variables in the convective boundary layer?, Bound.-Lay. Meteorol., 136, 253–284, https://doi.org/10.1007/s10546-010-9494-z, 2010.

Wyngaard, J. C.: The effects of probe-induced flow distortion on atmospheric turbulence measurements: Extension to scalars, J. Atmos. Sci., 45, 3400–3412, https://doi.org/10.1175/1520-0469(1988)045<3400:TEOPIF>2.0.CO;2, 1988.

Yu, Y., Liu, J., Chauhan, K., de Dear, R., and Niu, J.: Experimental study on convective heat transfer coefficients for the human body exposed to turbulent wind conditions, Build. Environ., 169, 106533, https://doi.org/10.1016/j.buildenv.2019.106533, 2020.